213 masthead unit plug and cable
Posted 09 December 2012 - 08:56 AM
So, question is, is this completely fucked, can the wire be replaced. How does one re-feed the individual wire through the grommet at the back of the plug? Or, do I need to buy a new plug and cable assembly.
Posted 09 December 2012 - 12:26 PM
If you need some new plugs or pins, you can order them at farnell/element14.
For instance the pins:
The hole unit is a standard AMP connector.
For the Vertical MHU I have the correct codes on my site:
For the horizontal I don't know the right order codes, so if you figure them out, please let us know.
Posted 10 December 2012 - 06:18 AM
It appears that the tapered rubber grommet at the back of the plug appears to be moulded around the wires once they are soldered in place. To get past this I have drilled this out to allow the wire to pass though and I will reseal with a silicon once everything is soldered in place. I think I have managed to save the pins which can be reused so hopefully everything will go back together and last another 10 years until it's someone else problem.
Posted 11 December 2012 - 08:48 PM
Posted 11 December 2012 - 10:25 PM
Posted 12 December 2012 - 10:33 AM
Posted 13 April 2014 - 05:06 PM
I 'm new on this forum so greetings to you all. I 'm also have a question about B&G mastunit 213. I removed the mastunit during the winter and when I replaced it something goes wrong. By replacing I turned it where the inside plug in the foot damaged and broke 5 of the 6 pins. Question is where can I find the connector to replace it. The connector is not a whole connector but only the print with the pins. At B&G Tinley they have no answer as it is an old unit.
I tried to place some pictures on this topic, but unfortunately I was not succeeded. In the helpfile on this topic I couldn't find how to do it so mayby one of you can give me some advice. Thanks for your help. Greetings from Rotterdam. Henk.
Posted 14 April 2014 - 07:14 AM
sorry to say, but you will probably have to replace the spar (part# 213-10-003).
Posted 15 April 2014 - 09:37 AM
On my last post I had a question about the 213 mastunit. By searching on the net I found out that my mastunit is not the 213 but the 283 wich is an unit used with the B&G Network. It was introduced in 1988 and discontinued at 1996, so it is an very old unit. The question is still the same. The connector is situated inside the spar. I have the whole unit dismanteled and I will replace the cables and the connector from wich the pins has broken. Where can I find a replaceconnector. Can I replace it with a connector from Farnell/Unit14? Thanks for help. Henk.
Posted 15 April 2014 - 09:57 AM
Yes, the tiny board can indeed be replaced... if you can find a spare one..
I checked my drawers and I only found a damaged thingie. However, I also found six new pins.... Ping me.
Posted 15 April 2014 - 10:39 AM
I posted some pictures about the connector 238. I posted from Photobucket and the pictures are very large. Sorry.
1 user(s) are reading this topic
0 members, 1 guests, 0 anonymous users